Background: Use of antigen-detecting malaria rapid diagnostic tests (RDTs) has increased exponentially over the last decade. WHO’s Global Malaria Programme, FIND, and other collaborators have established a quality assurance scheme to guide product selection, lot verification, transport, storage, and training procedures. Recent concerns over the quality of buffer packaging and test accessories suggest a need to include these items in product assessments. This paper describes quality problems with buffer and accessories encountered in a project promoting private sector RDT use in five African countries and suggests steps to avoid or more rapidly identify and resolve such problems.
Methods: Private provider complaints about RDT buffer vials and kit accessories were collected during supervisory visits, and a standard assessment process was developed. Using 100 tests drawn from six different lots produced by two manufacturers, lab technicians visually assessed alcohol swab packaging, blood transfer device (BTD) usability, and buffer appearance, then calculated mean blood volume from 10 BTD transfers and mean buffer volume from 10 individual buffer vials. WHO guided complaint reporting and follow-up with manufacturers.
Results: Supervisory visits confirmed user reports of dry alcohol swabs, poorly functioning BTDs, and non-uniform volumes of buffer. Lot testing revealed further evidence of quality problems, leading one manufacturer to replace
buffer vials and accessories for 40,000 RDTs. In December 2014, WHO issued an Information Notice for Users regarding variable buffer volumes in single-use vials and recommended against procurement of these products until defects were addressed.
Discussion: Though not necessarily comprehensive or generalizable, the findings presented here highlight the need for extending quality assessment to all malaria RDT test kit contents. Defects such as those described in this paper could reduce test accuracy and increase probability of invalid, false positive, or false negative results. Such deficiencies could undermine provider confidence in RDTs, prompting a return to presumptive treatment or reliance on poor quality microscopy. In partial response to this experience, WHO, FIND, and other project partners have developed guidance on documenting, troubleshooting, reporting, and resolving such problems when they occur.
- Populations Served
- Health Areas
- Developing Markets
- Resource Types
- Peer-Reviewed Publications
- Kenya, Madagascar, Nigeria, Tanzania, Uganda
- Fever Case Management (FCM), Malaria Rapid Diagnostic Tests (RDT), Screening, Diagnosis, and Treatment
- UNITAID Private Sector RDT Project